HIOKI 3284 MANUAL PDF
A Complete HIOKI Digital & Analog Clamp Tester /A AC .. CARRYING CASE ×1. Hand strap ×1. Instruction manual ×1. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual. View and Download Hioki instruction manual online. Clamp On AC/DC HiTester. Measuring Instruments pdf manual download.
|Published (Last):||20 January 2009|
|PDF File Size:||7.50 Mb|
|ePub File Size:||20.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
Replace the battery to ensure accuracy.
Hioki CLAMP ON AC/DC HiTESTER
Ask our experts To ask our Hioki experts a question please include your contact details. To print the manual completely, please, download it.
Just Added to Your Cart: Slide the slide cover up using the slide knob. Page 5 We have tried to bring this manual as close to perfection as we could achieve. Using a one-chip microprocessor, the Chapter 2 Measurement Procedure Explains how to use the for measurement. Shipping Check In particular, check the accessories, panel switches, keys, and terminals.
CLAMP ON AC/DC HiTESTER
The torque about 0. Ask an Expert More Hioki Products. We accept these payment methods: While the recording function is being deactivated, data is not updated, even if the clamp sensor is disconnected from the conductor.
However, hioii during use could result in injury or death, as well as damage to the instrument.
Using a Hioki Current Clamp Meter: Enter text from picture: Remove the rear cover and insert a battery. Our Member Store gives you access to the following advantages: Unit turns off during use. Also, when deactivating the recording function, press the key to terminate HOLD In the manual, the symbol indicates particularly important information that the user should read before using the instrument. Symptom Battery Battery clip Test leads Unit does not come indication appears and unit immediately turns off.
Then the model name is shown, and the bar graph indicates the battery condition. If this happens, perform a zero-cancel correction 2. Next, insert the red test lead to V and the black test lead to COM of the voltage measurement terminal.
Waveform and harmonic analysis. Request for Quote To request an official quote manal the Hioki please include your quantities and contact details. Auto-zero adjustment function and 2.
Select a proper mode according to the waveform shown below: Open gioki top ends of the clamp core and clamp the measured conductor so that it passes through the center of the clamp core. Therefore, you must select at least one lens with this item. These are defined as follows. A conversion table for ranges is provided below. Quit the peak measurement mode, and perform the auto-zero adjustment by.
Continue on ITM Canada. Always switch to the range you will use a manual range before performing the adjustment. Inrush current crest value. A smaller value will be displayed for a high inverter carrier frequency.
Hioki 3284 CLAMP ON AC/DC HiTESTER
Comments to this Manuals Your Name. By pressing ‘print’ button you 32844 print only current page. Don’t have an account? Indicates that incorrect operation presents an extreme hazard that could DANGER result in serious injury or death to the user.
Hioli the current range of the measured circuit is known, set the current range to the manual range. Call us toll free 1. This product includes one lens. Secondary electrical circuits connected to an AC electrical outlet through a transformer or similar device. Yes, Sign Me Up! Submit Request Add More Items. It is recommended that a new battery should be used, when the blackout seems to be short.
In particular, check the accessories, panel switches, keys, and terminals. Save my login information.